# Finding Distance Using Analog Voltage

### How to Calculate Distance from the Pin 3

The Analog Voltage pin on the MaxSonar family of sensors has been the most popular output for our users. All of the MaxSonar sensors have this output included. This guide will give a look into how to use it for obtaining the distance to the target being detected. The Analog Voltage output is scaled on all sensors to the power input of the sensor. The Analog Voltage output will output a linear voltage that gets larger as a target increases in distance from the sensor. This scaling will be outlined with each sensor family. For all examples the power input is +5VDC.

**Review the formulas below, organized by product line, to assist you in the calculating of distance.**

**If you have any questions, contact our Technical Support for assistance: [email protected]**

## Contents

1. LV-MaxSonar Sensors

1.1. Using an LV-MaxSonar with an ADC (Analog Digital Converter)

2. Standard XL-MaxSonar Sensors

2.1. Using a Standard Range XL-MaxSonar with an ADC (Analog Digital Converter)

3. XL-MaxSonar Long Range Sensors

3.1. Using a Long Range XL-MaxSonar with an ADC (Analog Digital Converter)

4. 5-Meter HR-MaxSonar sensors

4.1. Using a 5-Meter HR-MaxSonar with an ADC (Analog Digital Converter)

5. 10-Meter HR-MaxSonar sensors

5.1. Using a 10-Meter HR-MaxSonar with an ADC (Analog Digital Converter)

**LV-MaxSonar Sensors**

This is for the LV-MaxSonar-EZ, LV-MaxSonar-WR, and LV-MaxSonar-WRC. All of our LV-MaxSonar sensors range to targets in inches. The formula for the voltage scaling on an LV-MaxSonar sensor family is:

**[(Vcc/512) = Vi]**

Vcc = Supplied Voltage

Vi = Volts per inch (Scaling)

**Example 1**

[(5.0V/512) = 0.009766V per inch = 9.766mV per inch]

##### Calculating the Range

Once you know the voltage scaling it is easy to properly calculate the range.

The range formula is:

**[(Vm/Vi) = Ri]**

Vm = Measured Voltage

Vi = Volts per Inch (Scaling)

Ri = Range in inches

**Example 2:**

If your multimeter shows a measurement 292.98mV then you use the calculations as follows:

[(292.98mV/9.766mV) = 30 inches]

**Example 3:**

To work backward and verify your calculation is correct use the inverse formula:

**[(Ri x Vi) = Vm]**

[(30 x 9.766) = 292.98]

**Using an LV-MaxSonar with an ADC (Analog Digital Converter)**

When using an LV-MaxSonar with an ADC, verify that the sensor and micro-controller are referencing the same power supply and ground. This also assumes that the ADC being used is perfectly accurate.

When reading the sensor’s output with the scaling in inches with a 10-bit ADC, divide the ADC output by 2 for the range in inches.

If the ADC output reads 508 (maximum output) the range in inches is 254 inches. If the ADC output reads 250 the range in inches is 125 inches.

**Standard XL-MaxSonar Sensors**

This is for the XL MaxSonar EZ, XL-MaxSonar-AE, XL MaxSonar WR, and XL MaxSonar WRC. This formula does not include scaling for our 10 meter sensors: XL-MaxSonar-WRL1/WRLA1, XL-MaxSonar-EZL0/AEL0, and XL-MaxSonar-EZL1/AEL1.

The voltage scaling formula for the standard XL MaxSonar sensor family is:

**[(Vcc/1024) = Vcm]**

Vcc = Supplied Voltage

Vcm = Volts per cm (Scaling)

**Example 1:**

The formula should read:

[(5.0V/1024) = 0.004883V per cm = 4.883mV per cm]

##### Calculating the Range

Once you know the voltage scaling you can properly calculate the range.

The range formula is:

**[(Vm/Vcm) = Rcm]**

Vm = Measured Voltage

Vcm = Volts per cm (Scaling)

Rcm = Range in cm

**Example 2:**

If the multimeter shows a reading of 439.47mV, then you use the calculation as follows:

[(439.47mV/4.883mV) = cm] in this case the range is 90cm

**Example 3:**

To work backward and verify your calculation is correct use the inverse formula:

**[(Rcm x Vcm) = Vm]**

[(90 x 4.883mV) = 439.47mV]

**Using a Standard Range XL-MaxSonar with an ADC (Analog Digital Converter)**

When using a standard XL-MaxSonar with an ADC, verify that the sensor and micro-controller are referencing the same power supply and ground. This also assumes that the ADC being used is perfectly accurate.

When reading the sensor’s output with the scaling in centimeters with a 10-bit ADC, the range can be read directly off the ADC.

If the ADC output reads 700 the range in centimeters is 700 centimeters. If the ADC output reads 200 the range in centimeters is 200 centimeters.

**XL-MaxSonar Long Range Sensors**

For the 10meter long range sensors the calculation formula is

**[(Vcc/1024) = V2cm]**

Vcc = Supplied Voltage

V2cm = V per every 2 cm

Due to a limitation of hardware the Analog Voltage output steps in 2cm increments.

The formula to find the final distance is

**[(Vm/V2cm)=U] x 2= cm**

Vm = Volts measured

V2cm = V per every 2 cm

U = half measurement

**Example 1:**

Your multimeter has a reading of 439.47mV.

[(439.47/4.883)=90] x 2 = 180cm

**Example 2:**

To work backwards and verify correct calculation us the following formula

**[(cm/2) x V2cm]=Vm**

If you input the calculated range from Example 1 of 180cm into the formula, your reverse formula should read

[(180/2)x4.883]=439.47mv

**Using a Long Range XL-MaxSonar with an ADC (Analog Digital Converter)**

When using a standard XL-MaxSonar with an ADC, verify that the sensor and micro-controller are referencing the same power supply and ground. This also assumes that the ADC being used is perfectly accurate.

When reading the sensor’s output with the scaling in centimeters with a 10-bit ADC, the range output has a 2cm resolution.

If the ADC output reads 500 the range in centimeters is 1000 centimeters. If the ADC output reads 200 the range in centimeters is 400 centimeters.

**5-Meter HR-MaxSonar sensors**

**(Vcc/1024)=V5mm**

Vcc = supplied voltage

V5mm = Volts per 5mmDue to a limitation within the hardware of the sensor the Analog Voltage steps in 5‑mm increments.

**Example 1**

(5/1024)=V5mm or 0.004885V (4.883mV) per 5 mmAfter the voltage scaling has been found, range calculation is now able to be done.To calculate range use the formula:

**(Vm/V5mm)x5=Rmm**Vm = Volts Measured

V5mm = Volts per 5 mm

Rmm = Range in mm

**Example 2**

The multimeter being used shows a voltage of 2.441V (2441.4mV). To find the distance the formula would look like

(2441.4/4.883)x5=Rmm

For this example the Range in millimeters is 2500 millimeters

**Example 3**

To verify this is correct use the following formula:

**(Rmm/5) X V5mm)=Vm**So the reverse calculation would look like:

(2500/5)X 0.004883=Vm

After calculation the Vm should read 2441.4mV or 2.441V

**Using a 5-Meter HR-MaxSonar with an ADC (Analog Digital Converter)**

When using a 5-meter HR-MaxSonar with an ADC, verify that the sensor and micro-controller are referencing the same power supply and ground. This also assumes that the ADC being used is perfectly accurate.Using a 10bit analog to digital convertor, one can read the analog voltage bits (i.e. 0 to 1024) directly and just multiply the number of bits in the value by 5 to yield the range in mm. For example, 60 bits corresponds to 300-mm (where 60 * 5 = 300), and 1000 bits corresponds to 5000-mm (where 1000 * 5 = 5000-mm).

**10-Meter HR-MaxSonar sensors**

This is our newest line of sensors. To calculate the distance using voltage please use the following formula

**(Vcc/1024)=V10mm**

Vcc = supplied voltage

V10mm = Volts per 10mm

**Example 1**

(5/1024)=V10mm = 0.004883 in this case V10mm will be 0.004883V or 4.883mV.

After the voltage scaling has been found, range calculation is now able to be done.

To calculate range use the formula:

**(Vm/V10mm)x10=Rmm**

Vm = Volts Measured

V10mm = Volts per 10 mm

Rmm = Range in mm

**Example 2**

The multimeter being used shows a voltage of 1.2207V (1220.7mV). To find the distance the formula would look like

(1220.7/4.883)x10=Rmm

For this example the Range in millimeters is 2500 millimeters

**Example 3**

To verify this is correct use the following formula:

**(Rmm/10) X V10mm)=Vm**

So the reverse calculation would look like:

(2500/10)X 0.004883=Vm

After calculation the Vm should read 1220.7mV or 1.2207V

**Using a 10-Meter HR-MaxSonar with an ADC (Analog Digital Converter)**

When using a 10-meter HR-MaxSonar with an ADC, verify that the sensor and micro-controller are referencing the same power supply and ground. This also assumes that the ADC being used is perfectly accurate.

Using a 10bit analog to digital convertor, one can read the analog voltage bits (i.e. 0 to 1024) directly and just multiply the number of bits in the value by 10 to yield the range in mm. For example, 30 bits corresponds to 300-mm (where 30 * 10 = 300), and 1000 bits corresponds to 10,000-mm (where 1000 * 1 = 10,000-mm).

If you have any further questions or need assistance with sensor selection, contact our tech support team, click the button below!

### Sensors Related to this Article

## MB1003

HRLV-MaxSonar-EZ0

Features of the MB1003, HRLV-MaxSonar-EZ0, include millimeter resolution, the widest and most sensitive beam pattern of any unit from the HRLV‑MaxSonar‑EZ sensor line, short to long distance detection, range information from 300mm to 5000mm, a 10Hz read rate, and various output options: pulse-width, analog voltage, and a choice of either RS232 or TTL serial.

## MB1240

XL-MaxSonar-EZ4

Features of the MB1240, XL-MaxSonar-EZ4, include centimeter resolution, has the highest noise tolerance and narrowest beam width of any sensor in the XL‑MaxSonar‑EZ sensor line, short to long distance detection, range information from 20cm to 765cm, a 10Hz read rate, and various output options: pulse-width, analog voltage, and RS232 serial.

## MB7360

HRXL-MaxSonar-WR

Features of the weather resistant MB7360, HRXL-MaxSonar-WR, include millimeter resolution, short to long distance detection, range information from 300mm to 5000mm for the nearest detectable target, a 7.5Hz read rate, and various output options: pulse-width, analog voltage, and RS232 serial.